The Freedom V² CPAP Battery by Battery Power Solutions gives you the freedom to power your CPAP machine when and where you need it!
The Freedom V² features variable voltage output (12V-24V) combined with streamlined smart DC output cables with dynamic control technology to offer maximum efficiency and compatibility.
It is compact and lightweight making it great for use as a travel battery to power your CPAP machine while camping or on a red-eye flight. It can also be setup as a battery backup system to provide continuous power to your CPAP machine in the event of a power outage.
The Quick Charge 3.0 USB port lets you use the Freedom V² to charge your favorite portable electronic devices like cell phones, tablets, MP3 players, and more.
This Freedom V² CPAP Battery Kit includes the 150W Sine Wave Power Inverter which allows you to connect a variety of CPAP machines for which a specific Freedom V2 DC Output Cable is not available to provide 1-2 nights of power to these machines. If you need longer run times or you use a BiPAP machine, you can order a dual battery kit and bridge two Freedom V² batteries together for more power.
The 150W Sine Wave Power Inverter also lets you charge the Freedom V² CPAP Battery using the 12V auxiliary power outlet in a vehicle. Since this inverter is not proprietary to the Freedom V², you can use it with other external batteries and electronic devices.
Don't forget to check out the rest of our CPAP Supplies & Accessories designed to make using your CPAP machine easier and more comfortable!
*Dual battery kit will also include the Freedom V² CPAP Battery with Bridge Cable.
**CPAP machine NOT included.
The Freedom V² CPAP Battery is intended to provide external battery power to CPAP and other PAP devices when AC power is unavailable during travel or power outages. The Freedom V² battery can also be used to charge most common portable electronic devices such as cell phones and tablets.
Note: Use of the Freedom V² CPAP Battery with a device that has higher amperage requirements than the output of the battery (over 8A) may cause the internal circuitry of the battery to shut off to protect from thermal runaway.
|Cell Type:||Rechargeable lithium-ion (LG/Samsung ICR18650)|
|Output Voltage Range:||12V-24V DC (depending on DC output cable)|
|Output Current:||8A (maximum)|
|Average Run Times:|| Respironics DreamStation Go, Transcend and Z1/Z2 Travel CPAP machines =
Single Battery: 1-2 nights | Dual Batteries: 2-4 nights
Other PAP Device Models (i.e. 3B Medical Luna II) = Single Battery: 1 night | Dual Batteries: 2 nights
*night = 7 to 8 hrs
|DC Input Voltage Range:||10V-28V|
|QC 3.0 USB Output Voltage:||5V | 2.4A (maximum)|
|Optimal Charging Conditions:||32°F to 104°F|
|Optimal Operating Conditions:|| 23°F to 104°F
*battery capacity is reduced by 30% when operating between 23°F to 32°F
|Charge Time:|| 2-3 hrs @ 90W
*charge time is per battery and based on using the Freedom V² AC power supply
|Power Gauge:||Digital display screen (shows charge level as a percentage)|
|Storage Conditions:|| -4ºF to 104ºF
*optimal range is 60ºF to 80ºF
|Life Span:|| 500 charge/discharge cycles (100%-0% then back to 100%)
*after 500 cycles the battery shall recover at least 70% of its initial charge
|Dimensions:||7.5 " x 4.8" x 0.8" (L x W x H)|
|Warranty:||1 year limited|
Average Run Time Notes:
Many factors can affect how long the Freedom V² CPAP Battery battery will power your PAP device, including, but not limited to the device type (CPAP, BiPAP, AVAP, VPAP, etc.) pressure setting, your mask fit, and the use of heated humidification or heated tubing.
To increase average run times the Freedom V² CPAP Battery can be used in a dual battery configuration by bridging two Freedom V² batteries together.
Average run times are based on CPAP machines without the use of heated humidification or heated tubing on a pressure setting of 10 cm/H20.